Describe KAGRA/Subgroups/VIS/OpsManual/TypeB/ControlWork01 here.
- Estimation of the IP coupling ratio: L2T, L2Y, T2L, T2Y, Y2L, Y2T IP DC actuators diagonalization: measuring the DC coupling of the IP by pushing it in DC along L, T, Y Optical levers diagonalization method similar to type A one and fixing scale by moving the IP. For Pitch dof we will discuss later. TM actuators diagonalization: using the DC method like IP and checking the residual coupling by injecting whit noise or an high frequency line. Sensor correction: we need to modify the real time model and if it will be available soon we are gonna to implement SC Check the diagonalization of the geophone. Blending: we have to measure the noise of the sensor by looking at the open loop spectra of geophones and LVDT. IP control filters tuning