Differences between revisions 48 and 49
 ⇤ ← Revision 48 as of 2017-10-30 14:24:58 → Size: 4686 Editor: KokiOkutomi Comment: ← Revision 49 as of 2017-11-02 13:13:20 → ⇥ Size: 4695 Editor: KokiOkutomi Comment: Deletions are marked like this. Additions are marked like this. Line 10: Line 10: * ACC H1 (SN7): -5.288 [(um/s^2)/cnt] * ACC H2 (SN8): 5.830 [(um/s^2)/cnt] * ACC H3 (SN5): -5.289 [(um/s^2)/cnt] || || Calibration [(um/s^2)/cnt] || || ACC H1 (SN7) || -5.288 |||| ACC H2 (SN8) || 5.830 |||| ACC H3 (SN5) || -5.289 ||

Type-A Tower ETMX Specification

IP

Accelerometer (ACC)

|| || Calibration [(um/s^2)/cnt] ||

 ACC H1 (SN7) -5.288 ACC H2 (SN8) 5.83 ACC H3 (SN5) -5.289

LVDT

• LVDT H1: -0.258 [um/cnt]
• LVDT H2: -0.252 [um/cnt]
• LVDT H3: -0.233 [um/cnt]

Actuator polarization

Polarity is determined so that an +offset at the COILOUTF applies +yaw force in each actuator.

• COIL H1: negative
• COIL H2: negative
• COIL H3: positive

Diagonalization matrix

• ACC2EUL
 H1 H2 H3 To: 0.6440 -0.1726 -0.4714 Long. 0.1726 -0.6440 0.4714 Trans. 0.6833 0.6833 0.6833 Yaw
• LVDT2EUL
 H1 H2 H3 To: 0.6666 -0.3333 -0.3333 Long. 0 -0.5774 0.5774 Trans. 0.5612 0.5612 0.5612 Yaw
• EUL2COIL
 Long. Trans. Yaw To: 1. 0 0.5940 H1 -0.5000 -0.8660 0.5940 H2 -0.5000 0.8660 0.5940 H3

GAS

LVDT

• F0 LVDT GAS: -0.172 [um/cnt]
• F1 LVDT GAS: -0.215 [um/cnt]
• F2 LVDT GAS: -0.243 [um/cnt]
• F3 LVDT GAS: -0.225 [um/cnt]
• BF LVDT GAS: 0.204 [um/cnt]

Movable range by Fishing Rod

• F0 GAS:
• F1 GAS:
• F2 GAS:
• F3 GAS:
• BF GAS: full range (11 mm)

BF

LVDT (BF Damper)

• LVDT V1: -0.3785 [um/cnt]
• LVDT V2: -0.3852 [um/cnt]
• LVDT V3: 0.3902 [um/cnt]
• LVDT H1: 0.4060 [um/cnt]
• LVDT H2: -0.3821 [um/cnt]
• LVDT H3: -0.3629 [um/cnt]

Note: Polarity may change when connecting the secondary coil to the cable. This configuration is confirmed on [2017.10.27] (It will change after the cable replacement scheduled in November 2017.)

Actuator polarization

Polarity is determined so that an +offset at the COILOUTF applies +yaw force in each actuator.

• COIL V1: ???
• COIL V2: ???
• COIL V3: ???
• COIL H1: positive
• COIL H2: negative
• COIL H3: negative

Note: Polarity may change when connecting the secondary coil to the cable. This configuration is confirmed on [2017.10.27]

Diagonalization matrix

• LVDT2EUL
 V1 V2 V3 H1 H2 H3 To: 0 0 0 -0.5774 0 0.5774 Long. 0 0 0 -0.3333 0.6666 -0.3333 Trans. 0.3333 0.3333 0.3333 0 0 0 Vert. 0 1.5861 -1.5861 0 0 0 Roll 1.8315 -0.9158 -0.9158 0 0 0 Pitch 0 0 0 0.8170 0.8170 0.8170 Yaw
• EUL2COIL
 L T V R P Y To: 0 0 1. 0 0.3640 0 V1 0 0 1. 0.3152 -0.1820 0 V2 0 0 1. -0.3152 -0.1820 0 V3 -0.8660 -0.5000 0 0 0 0.4080 H1 0 1. 0 0 0 0.4080 H2 0.8660 -0.5000 0 0 0 0.4080 H3

KAGRA/Subgroups/VIS/TypeA/ETMX (last edited 2022-03-07 11:29:59 by hiromi.yasui)