|Deletions are marked like this.||Additions are marked like this.|
|Line 32:||Line 32:|
| * Calibrate SF LVDT.
* Install SF main ballast mass.
| * --(Calibrate SF LVDT.)--
* --(Install SF main ballast mass.)--
|Line 35:||Line 35:|
|* Add SF cap.||* --(Add SF cap.)--|
|Line 41:||Line 41:|
|* Install magic wand on PI.||* --(Install magic wand on PI.)--|
|Line 43:||Line 43:|
|* Rearrange magnets on SF damper magnet ring for full three Cu segments.||* --(Rearrange magnets on SF damper magnet ring for full three Cu segments.)--|
|Line 46:||Line 46:|
|* Lay more new in-air cabling.|
|Line 48:||Line 47:|
|* Get Cu O-rings for feedthroughs.||* --(Get Cu O-rings for feedthroughs.)--|
|Line 51:||Line 50:|
== Dates ==
* Repaired LBB half ring is returned: 9/11 (Mon, lunchtime)
* BUMAX countersunk screws for LBB Cu
* SDC screws for LBB half rings: 9/15 (Fri)
* LBB Cu segments: 9/19-20
|Line 73:||Line 78:|
|* Calibrate all LVDTs (Enzo, Mark). --(BF)--, SF, PI-GAS, PI-H1, PI-H2, PI-H3||* Calibrate all LVDTs (Enzo, Mark). --(BF)--, --(SF)--, PI-GAS, PI-H1, PI-H2, PI-H3|
VIS Type B BS Near Term To-Do List
- Vacuum and roll floor.
- Wipe ladders.
- Tidy tools, parts.
Get stuff from ../WishList
Main Assembly Sequence
Crane out SF.
Raise BF cap and retrieve lost hex cable clamp PEEK piece.
Crane in SF.
Install SF-BF rod with hex cable clamp.
Route cables via hex cable clamp up to below SF.
Adjust BF ballast mass for total weight and pitch/roll.
Look for missing screw from SF damper Cu segment in washed screws bag and reinstall.
Adjust yoke on SF to put top of secondary at 6 mm above top of baseplate (or to give ≈0 counts at nominal height).
Calibrate SF LVDT.
Install SF main ballast mass.
Cable and test SF fishing rod stepper.
Add SF cap.
- Lift and weigh SF/... section.
In parallel, time permitting
Rearrange arc weights on PI to allow installation of magic wand.
Install magic wand on PI.
- Check that we still have all LBB parts (there was an issue with borrowed lock nuts).
Rearrange magnets on SF damper magnet ring for full three Cu segments.
Find Ni-plated rods for SF damper magnet ring and do fit check.
Find PI-SF rod with 8 mm heads and matching receptacles and do fit check.
- Wipe LLB damper parts with ethanol.
Get Cu O-rings for feedthroughs.
- Figure out how to tighten PI yaw return spring.
Recalibrate remachined LBB springs and choose best three.
- Repaired LBB half ring is returned: 9/11 (Mon, lunchtime)
- BUMAX countersunk screws for LBB Cu
- SDC screws for LBB half rings: 9/15 (Fri)
- LBB Cu segments: 9/19-20
Tasks Not To Forget
- Update BS Install Procedure with screenshots from new on-assembly-frame 3D CAD.
- Finish assembly drawing for LBB and LBB damper ring and post on JGWDoc (Hirata-san).
- Get pico drive screens fixed (Miyo-kun)
- Calibrate PI vertical LVDT.
- Set yoke on PI vertical LVDT (if necessary).
- Calibrate PI horizontal LVDTs.
- Get clean bag laundered.
- Put pointy-tipped screw in RM trolley for BS
- Return/replace borrowed parts:
- Lock washers for LBB dampers
- Temporary ballast mass (dog clamps) in BF.
- Swap in new F0-SF maraging rod (with 8-mm heads) and new receptacles to match.
- Decide final in-vacuum and in-air cable routing (Enzo).
Calibrate all LVDTs (Enzo, Mark). BF, SF, PI-GAS, PI-H1, PI-H2, PI-H3
- Adjust yoke on PI to put top of secondary at TBD mm above top of baseplate (or to give ≈0 counts at nominal height).
Real-time and SUMCON models
- Create SUMCON models for full BS and major installation milestones.
Create controls design description (cf LIGO-T1100378).
- Make table of ADC/DAC assignments and check for consistency with D1503600 and k1visbs (Enzo, Mark)
- Understand/document/debug Kokeyama-san's DIO stuff.
- Measure LPCD and HPCD output filter response.
- Design and install software compensation filters for LPCD and HPCD output filters.
- Design and install GAS drift compensation filters.
Create a proper DAC Monitor MEDM screen.
Improve ADC Monitor MEDM screen.