|Deletions are marked like this.||Additions are marked like this.|
|Line 23:||Line 23:|
| * Get picomotor drivers back on line.
* Debug BS BIO connections, especially PI IP.
| * --(Get picomotor drivers back on line.)--
* --(Debug BS BIO connections, especially PI IP.)--
|Line 59:||Line 59:|
|* Move safety bars from BS AF area to SR2 area.||* --(Move safety bars from BS AF area to SR2 area.)--|
VIS Type B BS Near Term To-Do List
- Vacuum and roll floor.
- Wipe ladders.
- Tidy tools, parts.
Get stuff from ../WishList
Get picomotor drivers back on line.
Debug BS BIO connections, especially PI IP.
Main Assembly Sequence
- Necessary for phase-I★★★
Modify IP LVDT damping filters to include set point.
EXCHANGE YAW/VERT FRs at F0
- Necessary for phase-II★★
- Debug BS geophones.
- Flip circuit boards in BS geophones to make them the same as SRx.
- Design PI Sensor blending filter.
- Re-calibrate F0 GAS LVDT
- Tune the LVDT sensitivity
- Measure IP period; adjust ballast mass.
- At some point (remotely)★
- Improve OL offloading.
- Improve GAS offloading.
- Spectral analysis of Oplev.
- Design, install angular lock servo using OL. (Almost done for phase-I)
- Adjust gains and debug damping for IM OSEMs and GAS LVDTs. (Almost done for phase-I)
- Subtract DC level of Geophones.
- Spectra analysis of PI sensors (PI LVDTs and Geophones).
- Calibrate IP LVDT sensing against IP LVDT actuation, IP FRs.
- Set all EQ stops to TBD clearance.
- Modify the model for IP coil driver control
Documentation and Operation Manual
- Add useful info from Enzo's notebook to BSData.
Tasks Not To Forget
Move safety bars from BS AF area to SR2 area.
- Update circuit diagram.
- (Time permitting) Try adding mass to IP.
- Update BS Install Procedure with screenshots from new on-assembly-frame 3D CAD.
- Finish assembly drawing for LBB and LBB damper ring and post on JGWDoc (Hirata-san).
- Figure out how to make PI yaw adjustment reliable.
- Put pointy-tipped screw in RM trolley for BS
Return/replace temporary ballast mass (dog clamps).
- Characterize replacement F0.
Wipe/clean arc weights that were rained on.
- Update BS Circuit diagram, D1503600
Note removal of LED/PD on TM OSEMs.
- Organize stepper reset system and watchdog BIO.
- Include information on assignment of channels to feedthrough positions.
- Reflect gender-changers and non-flipping feedthrough adapters.
- Reflect actual cable lengths.
- Include LVDT pin size (1 mm).
- Include Stepper pin size (1.6 mm).
Undo renumbering of IP items.
Page 10: "DIO Card # 0" should be "DIO Card # 1"
Include OutConfigBoards on OL page (Sheet 18).
Update HPCD section to reflect:
All three GAS channels are on HPCD #1 and DIO Card #2, CNB cables, Sheet 13.
Three IP channels are on HPCD #0 and DIO Card #2, CNA cables, Sheet 15.
Real-time and SUMCON models
- Create SUMCON models for full BS and major installation milestones.
Create controls design description (cf LIGO-T1100378).
- Make table of ADC/DAC assignments and check for consistency with D1503600 and k1visbs (Enzo, Mark)
- Understand/document/debug Kokeyama-san's DIO stuff.
- Measure LPCD and HPCD output filter response.
- Design and install software compensation filters for LPCD and HPCD output filters.