|Deletions are marked like this.||Additions are marked like this.|
|Line 20:||Line 20:|
| * Shifted about 10cm in +X direction
* Pushed the NAB tower in +X direction about 10cm so that TCam's requirement can be fulfilled.
* With pushing it, lowered the NAB tower by the crane; large sound again, but it sat on the optical table somehow.
|Line 21:||Line 24:|
* Took out the inner tower from the IXA chamber.
* Was dangerous when it was hung up to float.
* Inserted the NAB suspension into the inner tower.
* Some screws were about to touch or going to touch the inner tower so we removed them.
* There were several points to be noted so that the NAB suspension won't crash against the inner tower, so be careful.
* Found everything are not fully consistent. Especially we needed to decide what should be used as reference. Today we used an array of through holes as the reference; the NAB suspension was somehow moved so that the interface through holes can overlap the screw hole array at the one side; the other side of the inner tower showed the through holes were totally useless to access the NAB suspension.
* Then measured several parts.
|Afternoon: Workers: Akutsu, Ikenoue, Simon, L. Shimizu
* Aligning the NAB tower further
Before noon: Workers: Akutsu, Ikenoue, Simon, L. Shimizu, Yokozawa
Hanged up the whole NAB tower (the NAB suspension + the inner tower) with the ceiling crane.
- Again laid the wooden (but covered by white tapes) bar to keep the distance of the two slings at the hanging points; this time the bar was softly taped to the inner tower's upper beam in order to avoid accidental fall off to the NAB.
Found the vertical move of the crane by one click was also ungentle.
Again used two (but new) koyoris to control or handle the NAB tower to avoid the accidental rotations (pitching, yawing, and rolling); the koyoris were handled by two workers.
Note: one crane person (Ikenoue) , two handlers (Simon and Shimizu), one watcher (Akutsu); so this work needs 4 people at least, hopefully 5 people (Yokozawa)!
- Transported on the IXA chamber.
- Handled by koyori as well.
- To find where is the center, firstly I stepped on the chamber and looked down at the center with my eyeballs.
- Hanged down the NAB tower into the IXA chamber while looking carefully whether the inner tower's four feet, which were the most sticking out, would not crash the chamber.
- Tried to sit down the NAB tower on the optical table in the chamber, but the one vertical step of the crane was so coarse (as described above) that it bounced on the table and made large sound!
- Hanged up the NAB tower again, slightly.
- I (Akutsu) entered into the chamber from the top, as the side doors were barriered by the inner tower's pillars; note that the NAB tower was hanged around the center of the chamber at this timing. First I tried to use a long ladder, which Uchiyama-san prepared, but we could not set the ladder stably in the chamber. So, we put a short ladder-like step, and I stepped down into the chamber with it from the top of the chamber.
- Then I put a laser leveler, and shot a vertical line so that the laser line was along a red line drawn on the optical table; the red line was drawn as a projection of the actual (but assumed) interferometer's laser line by Yokozawa-san, Uchiyama-san, and Telada-san the other day.
- Tried to set guide blocks along the feet of the NAB tower, when it was in the nice alignment and position. But we don't have nice ways today; today we used dog clamps, which were used to fix the inner tower's feet before, as the guide blocks, but we could not find how to fix them... anyway we used white tapes to fix them, but the tapes were too weak against the dancing NAB tower.
- Shifted about 10cm in +X direction
- Pushed the NAB tower in +X direction about 10cm so that TCam's requirement can be fulfilled.
- With pushing it, lowered the NAB tower by the crane; large sound again, but it sat on the optical table somehow.
Afternoon: Workers: Akutsu, Ikenoue, Simon, L. Shimizu
- Aligning the NAB tower further
- Chain block for 500kg. (What I found was that of 250kg, which Hirata-san had, and the NAB suspension +the inner tower would be comparable to this number so I felt it was dangerous to use the one.)
Guide blocks to guide the inner tower + NAB suspension. They are must!!
Modified dog clamps to clamp the legs of the inner tower. They are must!!
- A small bottle for ethanol
- Long ruler; use with a laser lever
- We found installing the dampers is mostly impossible the current structure; !!!! Urgently discuss how to resolve this!!!!
- We found the relative positions are not perfect and need to be resolved finally when the NAB would be suspended.
- Insert the NAB+tower into IXA
- Alignment and adjustment